One Spin Centrifuge
The simplest compact centrifuge, TOMY One Spin Mini Centrifuge accommodates one, two, three, or four tubes in any position without a counter balance. A mini-centrifuge designed for simplicity – close the lid to spin, release hand to stop.
- Can spin one tube without balancer
- Quick spin-down – Max 4,000 rpm
- Lid close to spin, release hand to stop
- Easy tube loading and unloading
- Compact design
- Spin column or screw cap tube compatible
One spin, one tube, one simple centrifuge.
Contact Us for customization and pricing.
Single phase, AC100-240V 50/60Hz, 0.6A | With AC adapter (6A-181WP05 with 5 types of plugs)
Spin a single tube
Auto-balanced TOMY One Spin Personal Centrifuge can spin a single 1.5 or 2.0 mL tube.
Personal Tabletop CentrifugeThe compact TOMY One Spin Personal Centrifuge can accommodate a single tube or up to 4 tubes. Excellent for use when only one tube needs to be spinned (no counter balance necessary). Includes rotors for 8 PCR tubes (1.5 / 2 mL Eppendorf-style) and 2 PCR strips (0.2mL x 8). Spin column or screw cap tube compatible.
One Touch SpinA simple, quick spin-down makes the TOMY One Spin Centrifuge a favorite amongst lab researchers. Load the tubes, then close the lid to spin. One touch operation, just hold the lid down with your hand, and release your hand to stop operation.
Capacity2.0 ml x 4 tubes
Maximum Speed4,000 rpm
Maximum RCF1,000 G
Power SourceAC or 4×AA batteries
|Max. Capacity||2.0ml × 4 microtubes|
|Tubes||2.0ml × 4 microtubes||1.5ml × 4 microtubes|
|RCF||450～1,000G (2.0ml)||420～950G (1.5ml)|
|Safety Devices||• Lid opening interlocking brake • Overcurrent fuse • Motor overcurrent detector (Motor lock, Unbalance)|
|Compliance of Standard||EU Directives • EMC Directive • Low Voltage Directive • RoHS Directive Reference Standard • EN61326-1 • EN61010-1 • EN61010-2-020|
|Dimensions||105W × 130D × 96H mm|
|Power Requirements||Dedicated AC adapter Input : 1-Phase AC100-240V 50/60Hz 0.6A|
|Power Consumption (calorific value)||2.0 kW (1720 cal/h)||3.0 kW (2580 cal/h)|
• AC adapter (6A-181WP05, 5 types of plugs) × 1set
• Operator's manual × 1 copy
Zero Installation Costs
Lifetime Customer Support
1 Year Warranty
What to Expect with
It’s easy to get your new autoclave set up.
Please ensure the following prior to receiving:
Measure the floor space for your new unit to make sure that your unit will fit perfectly.
Verify your existing electrical connections.
Your centrifuge requires 120V-grounded, 3-prong electrical outlet.
Empty & unplug your old unit.
Documents & Guides
Frequently Asked Questions
Q1. Are the TOMY MDX-310 Centrifuge rotors compatible with earlier model centrifuges?
Some centrifuge rotors are compatible with previous centrifuge models! It really depends on the rotors and which previous model centrifuge you have in your lab. We encourage you to reach out to our team to learn more. Contact us
Q2. Is the CA-300 rotor bucket a necessary accessory/component for operation?
Yes, the CA-300 rotor rack is essential for proper usage of the TOMY MDX-310 Centrifuge. It is what lets you stack up to 3 rotors in the centrifuge (depending on the specific rotor).
Q3. How does the TOMY MDX-310 Centrifuge know which rotor is currently inside the unit?
The TOMY MDX-310 Centrifuge is equipped with a Rotor Identification System. The system recognizes the installed rotor to operate within the optimal settings. The Rotor Identification System is an important safety feature that helps stop the unit from exceeding the maximum allowable speed for the specific rotor.
Q4. How do I load the tubes into the MDX-310 Centrifuge?
Like most centrifuges, the MDX-310 requires centrifugation tubes to be loaded symmetrically around the motor shaft. Opposing adapters, tubes, and samples should be of the same type and tubes should be filled equally to provide necessary counterbalance.
Q5. Does the MDX-310 Centrifuge have the ability to “flash” spin samples?
Absolutely! There is a FLASH function/mode and can even be set to Automatic Flash Mode. Details for short spins can be found in the MDX-310 Operator's Manual.
Q6. Can the MDX-310 display RCF as opposed to RPM?
Yes, adjusting the display to show RCF is simple settings change.
Q7. Can I store my favorite frequently used settings?
The MDX is equipped with a memory function that can save up to 6 sets of operating parameters including temperature, speed/RCF, run time and soft brake mode.
Q8. What is the best way to clean the rotors?
Remove the rotors, buckets, and racks from the unit and wash thoroughly with lukewarm (50°C) water, rinse and wipe dry. To sterilize use ethanol and avoid sterilizing in an autoclave or using dry heat.
Q1. Can this unit really run on battery power?
Yes, that is one of the key features of TOMY Multi Spin Centrifuge. The unit is fully operational on just 4 'AA' batteries. It can also run on AC power, and will give AC power priority if it is plugged in, so it won't drain the batteries. The battery and AC power variability makes the Multi Spin a great solution for "field" use.
Q2. Do I need to balance the tubes in the TOMY Multi Spin Centrifuge?
Yes, a balance loading pattern is required for proper operation of the unit.
Q3. What types of tubes can the Multi Spin handle?
The MA-1 rotor can utilize 2.0mL and 1.5mL tubes, while the PCR adapter (PA-1) holds 0.2mL tubes. Please refer to the specifications sheet for more detail.
Q4. What is included with the Multi Spin purchase?
The unit comes with the centrifuge, AC Adapter Set (5 types of plugs), the MA-1 Microtube rotor, the PA-1 tube stip rotor, and an Operator's Manual.
Q1. Can I really spin a single microtube in the One Spin without balancing?
Yes, this is the stand out feature for this particular centrifuge and is a great option for flashing a single microtube.
Q2. How do I operate the One Spin?
The centrifuge spins when the lid is closed. Load the tubes, close the lid and hold it down with your hand, release your hand to stop operation. It's that simple!